|Q:||How do I store my Columns?|
Recovering a partially dry column
To re-wet a partially dry column, install it in the reverse direction (to prevent the packing from damage). Using a flow of 0.1 ml/min, fill the column until no more bubbles appear at the column outlet. Switch the column to the correct flow direction and again use 0.1 ml/min for 1h. Then you can slowly increment the flow rate, systematically, using 0.2 ml/min steps.